The goal of this study was to determine whether the Tethapod system, which was designed to determine the impedance properties of lipid bilayers, could be used for cell culture in order to utilise micro-impedance spectroscopy to examine further biological applications. To that purpose we have used normal epithelial cells from kidney (RPTEC) and a kidney cancer cell model (786-O). We demonstrate that the Tethapod system is compatible with the culture of 10,000 cells seeded to grow on a small area gold measurement electrode for several days without affecting the cell viability. Furthermore, the range of frequencies for EIS measurements were tuned to examine easily the characteristics of the cell monolayer. We demonstrate significant differences in the paracellular resistance pathway between normal and cancer kidney epithelial cells. Thus, we conclude that this device has advantages for the study of cultured cells that include (i) the configuration of measurement and reference electrodes across a microfluidic channel, and (ii) the small surface area of 6 parallel measurement electrodes (2.1 mm2) integrated in a microfluidic system. These characteristics might improve micro-impedance spectroscopy measurement techniques to provide a simple tool for further studies in the field of the patho-physiology of biological barriers.
- Electrical impedance spectroscopy
- cell monolayer
- renal proximal tubule epithelial cell
- cancerous kidney cell
Over the last decade, the development of micro-dimensional systems to improve microscale measurements has attracted enormous attention due to their portability, high accuracy, and short response time. Consequently, they have evolved to a miniaturized automated version of biological measurements usually termed “Lab-on-Chip”. Impedance spectroscopy is a measurement technique that is very useful to characterize several electrical properties of materials and their interfaces with electrically conducting electrodes. It may be used to determine the dynamics of bound or mobile charges in the bulk or interfacial regions of any kind of solid or liquid material, including ionic, semiconducting, mixed electronic-ionic, and even insulators (dielectrics). Electrical measurements can evaluate the electrochemical behavior of electrode and/or electrolyte materials for living cells (1, 2). Indeed several devices have been developed over the last ten years that have minimized the number of cells for the analysis and also have reduced the response time to identify differences in cell behavior, such as monitoring of cell adhesion, spreading and motility of anchorage-dependent cells (3). Systems for impedance spectroscopy have also been designed for the analysis of single cells in suspension (4). Nonetheless, the barrier properties of epithelial cells are essential for their physiology and such measurements must be done on cell monolayers.
We have previously used the Tethapod system (SDX Tethered Membranes, Australia) to determine the impedance properties of lipid bilayers (5). The goal of this study was to determine whether this microdevice could be used for cell culture in order to measure impedance spectroscopy for further biological applications. To that purpose we have used normal epithelial cells from kidney (RPTEC) and a kidney cancer cell model (786-O).
Two types of epithelial cells from kidney were used in this study: the renal proximal tubule epithelial cells (RPTEC) and the tumour model of renal cell carcinoma (786-O cells). The normal RPTEC cells were purchased from (Evercyte GmbH) and grown at 37°C in ProXup medium. The 786-O tumour cells were purchased from Lonza and grown at 37°C in RPMI medium supplemented with fetal bovine serum (10%).
EIS measurements were performed using the Tethapod system (SDX Tethered Membranes, Australia). This system is usually used for EIS for measurements of the impedance properties of lipid bilayers (5). We used the T10 electrode chips that are supplied for use in the Tethapod. The T10 electrode chips had a pattern of gold electrodes that were supplied with a pre-coating of a stable monolayer that comprised a mixture of ester-free DLP (C49H92O11S2) and BnSS TEG (C15H24O4S2) molecules in the molar ratio of 10:90.
Renal proximal tubule epithelial cells (RPTEC) and renal cell carcinoma tumour cells (786-O) have been chosen in this study as they represent normal and tumour cells of the human kidney. Identifying distinct properties between normal and tumour cells is a major goal in cancer. These cells have been largely used in cell biology to study specific biological parameters such as cell proliferation, migration, and invasion as well as to determine specific potential targets to treat kidney cancer. The T10 electrode chip is provided in 100% ethanol. Prior to seeding of the cells the T10 electrode chips and the supplied microfluidic cartridge were incubated for 30 minutes in ethanol 70% to be adapted for cell culture. The T10 electrode chips and the microfluidic cartridge were dried thoroughly, and then the T10 electrode chip was assembled with the microfluidic cartridge (
The electrodes were then coated with fibronectin (7 μg/mL) for 30 minutes at 37°C using the entry ports of the microfluidic cartridge. After rinsing the fibronectin with culture media 10,000 cells of each type were seeded onto 5 wells of a T10/microfluidic cartridge in 200 μL of culture medium, and culture medium alone (no cells) was added to the 6th well. The T10/ microfluidic cartridge was incubated at 37°C and 5% CO2 for 3 days, with the durations indicated in text and figure legends.
As shown in
To perform the EIS measurements using the Tethapod, the T10/ microfluidic cartridge was removed briefly from the incubator after 1, 2 and 3 days. On the second day, after the EIS measurement, fresh medium was added to each well after carefully aspirating the old medium. The experiments were repeated on 3 separate occasions with different T10/microfluidic cartridges. With 5 electrodes used in each T10/microfluidic cartridge, the experiments were performed over 15 separate cell monolayers for each type of cell.
The total impedance signal was modelled using the equivalent circuit shown in
The results for the EIS measurement are presented in
Over the period of the 3 days, the capacitance of both the RPTEC and 786-O cell monolayers did not change (F = 0.053, p = NS). However, the capacitance of the RPTEC cell monolayer was lower than that of the 786-O cell monolayer (F = 12.912; p = 0.001). There was no significant interaction effect between the time of measurement and the cell-type (F = 0.237, p = NS). At the frequency range of the EIS measurements this capacitance is a combination of the individual capacitances of the apical and basolateral membranes of the cells in the monolayers. The difference in capacitance between the RPTEC and 786-O cell monolayers suggests that the membrane properties were also different between the RPTEC and 786-O cells.
We report here that the Tethapod, which is normally used for EIS measurements on tethered lipid bilayers (6), can be useful for assessing biological changes in cell cultures. The Tethapod is a swept frequency ratiometric impedance spectrometer for low-voltage (20 mV) AC impedance spectroscopy measurements over the frequency range from 0.125Hz to 1000Hz. Indeed, the Tethapod provides three advantages for the EIS measurements on the cell monolayers. Firstly, the large ground electrode faced the much smaller measurement electrode across a closed microfluidic channel of 100μm, which yielded homogeneous electric fields that were less sensitive to variations in the vertical cell position as compared to a configuration of coplanar electrodes on the same side of the channel (7). Secondly, the measurement electrodes have a small surface area (2.1 mm2) and are included within an integrated microfluidic system that comprised 6 electrodes. This allowed 6 separate cell monolayers and/or control conditions to be measured simultaneously. Each electrode is accessed by separate microfluidic channels that facilitates the convenient seeding of the cells, the growth of the monolayer and the convenient removal of the media for sustained cell culture. Thirdly, in the frequency range of the Tethapod the paracellular resistance, between the cells, and the capacitance of the cell monolayer predominately contribute to the total impedance (8). In this frequency range the large cell capacitance is predominate in the current flow across the cell, and we can summate the apical and basolateral membranes as one capacitance (cell monolayer) that we use in modelling the EIS measurements.
The lower resistance that we measured for the cancerous (786-O) cell monolayer compared to the normal (RPTEC) cell monolayer suggested that the cancerous cell monolayer had an increased permeability compared to the normal cells. This would suggest that the barrier properties of the cancerous cell monolayer were reduced, which is probably not surprising given that the growth of cancerous cells is usually disordered (9). The higher capacitance that we measured for the cancerous (786-O) cell monolayer compared to the normal (RPTEC) cell monolayer suggested that the membranes of the cancerous cell monolayer were different to those of normal cells. This measured capacitance was a combination from the capacitances of the apical and basolateral membranes, but nonetheless the measured capacitance suggested a difference in either or both of these membranes in the cancerous cells. The frequency range for our EIS measurements precluded the influence of cytoplasm properties, since it has been reported that the cytoplasm conductivity becomes prevalent at frequencies approaching 10 MHz (10). The difference in capacitance that we measured could indicate, for example, differences in the composition of the phospholipids or the composition of membrane proteins that could have altered the dielectric properties of the cell membrane (11, 12, 13). The measurement of capacitance is sensitive to changes in the dielectric properties, and hence our measurements of capacitance may provide a measurement of the difference in composition of the membranes of cancerous cells compared to normal cells. Albeit for a different purpose, the specific membrane capacitance was shown to change during the differentiation of neural stem cells, which was due to changes in the expression of membrane proteins (14). It is likely that changes in the composition of the membranes of normal cells becoming cancerous may similarly induce changes in the dielectric properties of the cells. With this Tethapod system we were able to measure simultaneously both the differences in membrane properties of cell monolayers of cancer compared to normal cells, which may provide insights into the changes in composition of the membranes of these cells.
Our results of decreased resistance but increased capacitance for cancerous kidney epithelial cell monolayers has been previously observed in another cell-type. In that report, the authors found similarly to our measurements that the resistance of the normal skin cell monolayer (HaCaT) was higher than the cancerous cell monolayer (A431), but that the capacitance of the normal cell monolayer was quite similar to the cancerous cell monolayer (3). The measurements that we report here might also provide information to further use these micro-EIS measurements for investigating the physiology of the adhesive properties of epithelial kidney cancer cells. Indeed, as the function of the kidneys depends on the maintenance of an epithelial barrier, the paracellular permeability to solutes and water might be modified in cancer. Thus, membrane properties of cancer cells identified by our measurements could lead to further investigations proteins known to have a major role into tight junctions between cells (15). As an example, claudin-2 a protein from tight junctions is highly expressed in kidney, liver, pancreas, stomach, and small intestine with a highest level in kidney. The presence of claudin-2 causes the formation of cation-selective channels that then increase the permeability between cells, whereas claudin-1 and claudin-4 decrease the permeability of the tight junctions (16). Thus, this device can be thereafter dedicated to study the specific role of one protein that could be incorporated in lipid bilayers.
We report that the Tethapod system, which was designed to determine the impedance properties of lipid bilayers, had a small measurement electrode surface area over which we could grow a monolayer that remained functionally viable for several days. To that purpose we have used normal epithelial cells from kidney (RPTEC) and a kidney cancer cell model (786-O). We demonstrate that the Tethapod system is compatible with the culture of cells 10,000 cells seeded for each of 5 measurement electrodes for simultaneous micro-EIS recordings. Furthermore, the range of frequencies for EIS measurements were tuned to examine easily the characteristics of the cell monolayer, including that we demonstrate significant differences in the paracellular resistance pathway between normal and cancer kidney epithelial cells. Thus, we conclude that this device has advantages for the study of cultured cells that include (i) the configuration of measurement and reference electrodes across a microfluidic channel, and (ii) the small surface area of measurement electrodes (2.1 mm2) integrated in a microfluidic system. These characteristics might improve micro-impedance spectroscopy measurement techniques to provide a simple tool for further studies in the field of the patho-physiology of biological barriers.